- Tripp Lite
- Ships in 24 Hours
- 4 Power Outlets
- Power NEMA 5-15 (F) Power NEMA 5-15 (M) Power Connectors
- Warranty: Limited Lifetime
Offering 15A AC power distribution for 120V applications in a rugged all-metal case, Tripp Lite's PS-415-HGULTRA is agency approved for use in a variety of healthcare venues, including physicians' offices, imaging centers, laboratories and more. Featuring internal site-wiring fault protection, it resets automatically with no need for action by the user. Features include switchless design to prevent accidental shutoff, four NEMA 5-15RHG hospital-grade outlets with safety covers, NEMA 5-15PHG hospital-grade plug, 15A resettable circuit breaker and optional-use keyhole mounting slots and flanges.
Approved for use in Patient Care Areas
Rugged all-metal housing stands up to the most demanding environments
(4) NEMA 5-15R-HG outlets with safety covers
15ft cord with NEMA 5-15P-HG plug
Switchless design prevents accidental shutoff
15A circuit breaker
Multiple installation options with integral mounting flanges and slots
UL 60601-1 and UL 60950-1 compliant
U.S. Patent No. 7,375,939
Suitable for use where Article 517 of the NEC requires Hospital-Grade Outlets
AC power distribution with hospital-grade receptacles and plug in healthcare applications where durability and reliability are required, including areas where NEC requires hospital-grade receptacles
Patient rooms, treatment rooms, exam rooms, etc.
TUV tested to UL 60601-1 and UL 60950-1 (USA); TUV tested to CAN/CSA-C22.2 No. 601.1-M90 and 60950-1-03 (Canada)
# Ports: 4Connector: Power NEMA 5-15 (F), Power NEMA 5-15 (M)
Color: WhiteForm Factor: Strip
1.5"h x 13.75"w x 2.5"d
(4) NEMA 5-15R-HG outlets
(1) NEMA 5-15P-HG input cord, 15ft.
Voltage compatibility: 120VAC
Frequency compatibility: 50/60Hz
Recommended Electrical Service: 15A, 120V
This product is subject to our return policy. Please see our complete complete return policy for details.
Compliant Standards: CAN/CSA C22.2 No. 601.1-M90, CAN/CSA C22.2 No. 60950-1-03, TUV, UL 60601-1, UL 60950-1